PYR2-420 – Class C Thermopile Pyranometer
PYR2-420 is an ISO 9060 Class C (Second Class) Thermopile Pyranometer.
It has a simpler construction than the Class B, but uses the same high quality materials.
As the Class C is equipped with the same electronics as the Class B, the signal has a high level of linearity and noise rejection. The signal is normalized in the 4 to 20mA range.
This analog output mode of the signal (current loop) is characterized by a simple wiring (cable with 2 wires), and a high immunity to electromagnetic interference and interference caused by distances of over ten meters.
This instrument is suitable for measuring solar power and monitoring the efficiency of photovoltaic systems when a precision of 5% – 10% is sufficient.
Note: the terms solarimeter, irradiance sensor, pyranometer and calibrated cell are interchangeable. On this site we will use irradiance sensor. To learn more click here.
Key features of a Class C Thermopile Pyranometer
- Thermopile sensor
- Double dome in hemispherical optical type glass with high transparency
- Anodized aluminum body ideal for outdoor use
- Output on RS485 modbus line, f.s. 1600W/ m2 (± 2W) for distances up to 500m
- 50m current loop cable
- High weather resistance
- ISO 6090 compliant
- Accurate calibrations with a Secondary Standard (Class A Pyranometer) rechecked every 12 months.
|PRODUCT||Class C Thermopile Pyranometer|
|REFERENCE STANDARD||ISO 9060:1990, ISO 9060:2018|
|OUTPUT||4-20 mA current loop|
|Calibration||ISO9847 compiant||With a Secondary Standard Pyranometer, certified calibration|
|Prectral Sensitivity||300-2900 nm|
|Input range||Irradiance||0-1600 W/m2|
|Response time||< 25 seconds|
|Resolution||Smallest change detected||± 8W/m2|
|Working temperature||-40 to +80°C|
|Output||Current Loop||4-20mA normalized output|
|Precision||Tilt response angle (0-90°)||± 4%|
|Response temperature ( ∆t = 50°K)||< 8%|
|Supply||9-30v DC protected against short circuit|
|Encapsulation||Quarzo [k5] dome||Optical glass with elevated transparency: 0.3-3 μm|
|Connectors||standard M8 3 pin female|
|Dimensions||162mm diameter x 104mm height|
(1) recommended recalibration within 20-24 months to maintain the specifications declared in calibration